Fast Veterinary Pregnancy Test Kit , Cattle Pregnancy Test Kit 5
Cattle Pregnancy Rapid Test Cassette is a lateral flow
immunochromatographic assay for the qualitative detection of
progesterone (P4) in cattle’s plasma or serum or milk.
Progesterone(P4) is a specific marker for indicating the activities
of cow ovary. It plays an important role in the process of maternal
reproduction and maintaining pregnancy.
|Production Name:||Cattle Pregnancy (Progesterone) Test VIPROG-202 5 Minutes
VETERINARY Bovine Range||Principle:||Milk/Plasma/Serum|
|Reading Time:||5 Minutes||Storage Temperature:||4-30℃|
|Accuracy:||94.40%||Shelf Life:||2 Years|
Easy visually interpretation
Simple operation, no equipment required
Cattle Pregnancy Rapid Test Cassette (Plasma/Serum/Milk) is a
lateral flow immunochromatographic assay for the qualitative
detection of progesterone (P4) in cattle’s plasma or serum or milk.
Progesterone (P4) is a kind of steroid hormone. The concentration
change of P4 is one specific characteristic indicating the
activities of cattle ovary. It plays an important role in the
process of maternal reproduction and maintaining pregnancy.
Therefore, it can be monitoring cattle reproductive activity by
detecting the concentration of P4. P4 concentration has different
characteristic changes as the cattle's stage of reproductive
physiology, through to detect the progesterone levels can make an
accurate diagnosis of cattle in early pregnancy.
INTERPRETATION OF RESULTS
Positive: If both T line and C line are visible, regardless of T line being
strong or faint, the test result is positive, which means the
cattle is pregnant or in middle period of estrous cycle.
Negative: If only C line is visible, the test result is Negative, which
means the cattle is not pregnant.
Invalid: When control line does not appear on the membrane, the test is
invalid since improper test procedure or deterioration of reagents
probably occurred. It is recommended that the test would be
DIRECTIONS FOR USE
Allow the test cassette, specimen and/or controls to equilibrate to
room temperature (15-30 °C) prior to testing.
1. Remove the test cassette from foil pouch and place it on a clean
and level surface
2. Hold the dropper vertically and transfer 4 drops of sample
(approximately 100 μl) into the specimen well(S), then start the
timer. See illustration below.
3. Wait for the colored line(s) to appear. Read results in 5
minutes. Do not interpret results after 10 minutes.